Analysis of the dead zone is among the intensive studies in a closed-loop fiber optic gyroscope. In a dead zone, a gyroscope cannot detect any rotation and produces a zero bias. In this study, an analysis of dead zone sources is performed in simulation and experiments. In general, the problem is mainly due to electrical cross coupling and phase modulation drift. Electrical cross coupling is caused by interference between modulation voltage and the photodetector. The cross-coupled signal produces spurious gyro bias and leads to a dead zone if it is larger than the input rate. Phase modulation drift as another dead zone source is due to the electrode contamination, the piezoelectric effect of the substrate, or to organic fouling. This modulation drift lasts for a short or long period of time like a lead-lag filter response and produces gyro bias error, noise spikes, or dead zone. For a more detailed analysis, the cross-coupling effect and modulation phase drift are modeled as a filter and are simulated in both the open-loop and closed-loop modes. The sources of dead zone are more clearly analyzed in the simulation and experimental results.
© 2015 Optical Society of AmericaFull Article | PDF Article
OSA Recommended Articles
Yue Zheng, Chunxi Zhang, Lijing Li, Lailiang Song, and Wen Chen
Appl. Opt. 55(17) 4589-4595 (2016)
Appl. Opt. 50(25) E152-E161 (2011)
M. Komachiya, H. Sonobe, S. Oho, K. Ohbu, T. Yuhara, and H. Iizuka
Appl. Opt. 35(19) 3719-3725 (1996)